In this post, I’m going to discuss a specific test for evaluating the resolution or lineshape of a benchtop NMR system. This test is measured on proton (1H) as the NMR spectrum is sensitive to the spectrometer resolution and we can make the measurement with a single scan. Resolution and lineshape refer to the width of a particular NMR spectral line, measured at 50% and 0.55% of the height of the line, as explained below. The smaller the linewidth value, the better the resolution.
1H lineshape and resolution
The information content of any NMR spectrum depends on the ability to observe and resolve different signals, or peaks, in the spectrum. It is easier to distinguish two sharp (narrow) peaks close together, than two broad peaks. The key technical factor defining the sharpness of the lines in the spectrum (the lineshape) is the homogeneity of the magnetic field generated by the magnet. Although NMR systems utilize an array of coils (“shims”) to further improve the B0 homogeneity, the achievable lineshape and resolution are strongly influenced by the inhomogeneity of the magnet itself. The process of calibrating the field and optimizing the B0 homogeneity is usually referred to as shimming. The homogeneity and resolution will gradually degrade over time, so you should carry out the shimming procedure whenever you want to ensure you have the best resolution your spectrometer is capable of. You should also do the shimming procedure immediately before making the linewidth test described here.
Measuring 1H lineshape and resolution
The way to measure instrument resolution is to collect a spectrum containing a naturally sharp line. The standard approach for is to use a sample containing chloroform in acetone-d6 – the chloroform signal from this sample has a very narrow natural linewidth.1 Figure 1 shows a 1H NMR spectrum sample of a 20% chloroform in acetone-d6 NMR reference sample collected on an 80 MHz benchtop NMR spectrometer. (https://www.sigmaaldrich.com/catalog/product/sial/611859)
Figure 1. 80 MHz 1H NMR spectrum of 20% chloroform in acetone-d6 (“lineshape”) sample